|Alternative Title||Study on High-throughput Biological Oxygen Demand Optic and Electrochemical Combination Detection Based on Polyaniline|
|Place of Conferral||北京|
|Keyword||生化需氧量 光电复合 聚苯胺 高通量|
生化需氧量，简称 BOD，指的是微生物分解水中有机物所消耗的氧气，是反映水体污染程度的一个重要指标。目前 BOD 检测方法主要有五日检测法和微生物电极法。五日检测法耗时长，重复性差；微生物电极法虽然检测迅速，但是存在着信号漂移问题。为了满足快速准确检测 BOD 的需求，研究人员发展出了光电复合检测 BOD 的方法。该方法将全内反射椭偏光学成像系统和电化学工作站相结合，利用聚苯胺修饰电极，实现了光信号和电信号的同步输出，解决了单一信号输出时存在的信号漂移问题。
Biochemical oxygen demand (BOD) is an important indicator for characterizing water pollution. It refers to the oxygen consumed by microorganisms to decomposing organic matter in water. The methods for estimating BOD include five-day BOD test and BOD biosensor based on an oxygen electrode. The five-day BOD test needs a long time and has poor repeatability; the BOD biosensor based on an oxygen electrode detects rapidly, but it has the problem of signal drift. In order to realize the needs of estimating BOD fast and accurately, an optic and electrochemical combination method has been developed. This method combines total internal reflection imaging (TIRIE) ellipsometry with electrochemistry to obtain the optic and electrochemical signals simultaneously by using polyaniline modified electrodes.
For the current combination system, there are still two problems: (1) The optical response mechanism of the polyaniline film is not clear, and the relationship between the optical signal and the electrical signal is also unclear; (2) The combination system uses a single-channel reaction cell, which cannot take advantage of the high-throughput detection of TIRIE system.
In order to solve the above problems, on the one hand, we deduce the relationship between the optical signal and the proportion of the oxidized state of the polyaniline film of TIRIE system based on the change of the optical thickness during the redox process of the polyaniline film, and use the Nernst equation to obtain the relationship between the electrochemical signal and the proportion of the oxidized state of the polyaniline film. The relationship between the optical signal and the electrochemical signal is obtained, and the optical response mechanism is analyzed. On the other hand, we designed a dual-channel and four-channel reaction cell, and completed the high-throughput detection of BOD by using the designed reaction cell, which greatly improved the speed of the combination system.
|马可. 基于聚苯胺的高通量生物需氧量光电复合检测研究[D]. 北京. 中国科学院大学,2019.|
|Files in This Item:|
|马可_基于聚苯胺的高通量生物需氧量光电复（3380KB）||学位论文||开放获取||CC BY-NC-SA||Application Full Text|
|Recommend this item|
|Export to Endnote|
|Similar articles in Google Scholar|
|Similar articles in Baidu academic|
|Similar articles in Bing Scholar|
Items in the repository are protected by copyright, with all rights reserved, unless otherwise indicated.